[Date Prev][Date Next][Thread Prev][Thread Next][Date Index][Thread Index]
Re: SiRF module and bluetooth
Gary E. Miller
Re: SiRF module and bluetooth
Fri, 26 Jun 2020 19:56:38 -0700
On Fri, 26 Jun 2020 19:37:11 -0700
Dave Riesz <firstname.lastname@example.org> wrote:
> Well, I think I may bail on the bluetooth link since it's adding
> complication without giving me much in return.
> Part of your response gets to what I'm working on that doesn't deal
> directly with gpsd: signals and noise in the receiver.
Easy to get lost yak shearing.
> A while ago I wanted to try to use a pair of GPS receivers to collect
> some reasonably accurate (or at least consistent) position
> information for my property using one unit as a stationary reference
> and the other to collect locations.
SiRF III does not fall in the "reasonable accurate" bucket.
> The units are identical. Same
> receiver module, same antenna, same RPi hardware, same software.
> Yet, even with the antennas sitting side-by-side the units would
> report positions that not only were quite different, but also jumped
> around with apparent randomness.
Yup. This is why gpsprof was invented. Give it a shot, you will
be surprised by the results.
SiRF claims 2 meter "accuracy". They define "accuracy" as what you get 50%
of the time. The rest of the time, anything goes.
> Thus my attention turned to noise
> in the system. I'd read that the 3.3v pins on the RPi (that I was
> using to power the GPS modules) weren't very clean
Yes, nut not in a way that affects GPS.
> and that the local
> oscillator(s) on a GPS unit could be sensitive to voltage
No. But they are sensitive to temperature. Once the receiver has
sat lock, that goes away.
> So I tried an RF choke on the power leads and then a
> separate power supply. The bluetooth idea was to totally isolate the
> receiver from the RPi.
BT is 2.4GHz, so the worst noise source you have. Not that it matters
> Anyway, if noise from the computer really isn't an issue, what might
> be causing the discrepancy between the receivers? The location and
> setup of my test units is not ideal and checks all of the boxes you
If you really want to understand the actual error sources, you need
to devote a lot of time to the Stanford video course on GPS:
For starters, the GPS ephemeris is only good to about +/- 1.5 m. Then
add in effects from ionosphere, troposphere, atomic clocks, etc.
> - limited skyview: right next to the exterior house wall
> - multipath reflection: house is wrapped in a steel stucco lath
> - cable loss: cheap mouse antenna probably has a lossy cable
> Not a great setup, but again both units are identical and are seeing
> very different things.
Can you define "very different". 20 meters would not surprise me.
Better yet, run a 2 hour gpsprof plot and send it here.
First thing: get a u-blox 9-series and your life gets much better.
Gary E. Miller Rellim 109 NW Wilmington Ave., Suite E, Bend, OR 97703
email@example.com Tel:+1 541 382 8588
Veritas liberabit vos. -- Quid est veritas?
"If you can't measure it, you can't improve it." - Lord Kelvin
Description: OpenPGP digital signature